arXiv:0906.1144v1 [physics.ins-det] 5 Jun 2009

Test of a LYSO matrix with an electron beam between 100 and 500 MeV for KLOE-2

M. Cordelli^a, F. Happacher^a, M. Martini^a, S. Miscetti^{a,∗}, I. Sarra^a, M. Schioppa^b, S. Stucci^b, G. Xu^c

aLaboratori Nazionali di Frascati dell’INFN, Frascati, Italy bINFN and department of physics, University of Calabria, Cosenza,Italy cInstitute of High Energy Physics of Academia Sinica, Beijing, China

Abstract

The angular coverage extension of the KLOE-2 electromagnetic calorimeter, from a polar angle of 20^∘ down to 8^∘, will increase the multiphoton detection capability of the experiment enhancing the search reach for rare kaon, η and η^′ prompt decay channels. The basic layout of the calorimeter extension consists of two small barrels of LYSO crystals readout with APD photosensors aiming to achieve a timing resolution between 300 and 500 ps for 20 MeV photons. The first test of a (5.5×6×13) cm^3 prototype for such a detector was carried out in april 2009 at the Beam Test Facility of Laboratori Nazionali di Frascati of INFN with an electron beam from 100 to 500 MeV. In the selected energy range, we measured a light yield of 500÷800 p.e./Mev, an energy resolution which can be parametrized as 0.05 ⊕ 0.01/(E/GeV) ⊕ 0.015/ √ E/GeV, a position resolution of 2.8 mm and a timing resolution of 200÷300 ps.

Key words: Calorimetry, LYSO, KLOE-2, timing resolution PACS: 29.40.Vj, 29.40.Mc

1. Introduction

In the last years, a new machine scheme based on the Crab-waist and a large Piwinsky angle has been proposed and tested [1] to improve the reachable luminosity at the Frascati ϕ-factory, DAΦNE, an e^+e^− collider running at the center of mass energy of the ϕ resonance. The success of this test motivated the startup of a new experiment, named KLOE-2 [2], which aims to complete the KLOE [3] physics program and perform a new set of interesting measurements. The first running phase will start at the end of 2009 with the goal of collecting ∼ 5 fb^{−1} in one year. For this first run, the modification of the KLOE detector will be minimal with the only introduction of a γγ tagging system. A second phase, for a longer data taking, will require another set of upgrades all concentrated around the beam-pipe consisting of: (i) an inner tracker, IT, based on Cylindrical GEM technology, (ii) a tile calorimeter surrounding the inner quadrupoles , QCALT , and (iii) a calorimeter between the interaction point, IP, and the first inner quadrupole. In the following, we describe the work done to design and test the possibile solution for the latter detector.

In Fig. 1, we show a zoomed-view of the available region around the IP which can be used to extend the angular coverage of the main electromagnetic calorimeter, today limited at a polar angle of ∼ 20^∘, with the addition of a new dedicated calorimeter. Assuming to be able to lower the minimum polar angle for photon detection down to 8^∘, this will enhance the multiphoton detection capability of the detector for the search of rare decays of kaons, η and η^′ mesons. As an example, in 2007 the KLOE experiment has measured a BR(K_S → γγ) [4]

∗Corresponding author Email addresses: Stefano.Miscetti@lnf.infn.it (S. Miscetti)

Figure 1: Zoomed-view of the IP region. The area available for the new calorimeter lies between the inner sphere and the closest quadrupoles.

in good agreement with O(p^4) calculation of Chiral Perturbation Theory but which differs of three standard deviations from the same measurement carried out by the NA48 experiment. The proposed upgrade will allow to improve the S/B ratio of a factor of three thus allowing to solve this puzzle. Other searches or branching ratio determinations, such for instance K_s → 3π^0 or η → π^0γγ, will directly benefit from this extension. The only available area to place a calorimeter lies between the end of the spherical beam pipe, of 10 cm radius, and the first quadrupole, positioned at 30 cm distance from the IP.

2. CCALT: a Crystal Calorimeter with Time

The discussion of the previous section indicates that this calorimeter has to be very dense, with a small value of radiation length, X_0, and Moliere radius, R_m, not hygroscopic and with a large light output to improve photon detection efficiency at low energy (from 20 to 500 MeV). Moreover, the calorimeter has to be extreemely fast in order to allow prompt photon

Preprint submitted to Elsevier May 5, 2019

Figure 2: Composition of crystals used for the matrix prototype.

reconstruction in an environment with a large background rate (∼ 1÷ 5 MHz) of secondary showers generated by off-axis e^+, e^− coming from intra-bunch scattering (Touschek effect). Preliminary simulation studies indicates the need to reach a time resolution of 300÷ 500 ps for 20 MeV photons.

A suitable solution is offered by a crystal calorimeter with good timing performances, named CCALT. A first detector layout consists of two concentrical barrels of 24 crystals each, with transversal dimensions of 2×2 cm^2 and longitudinal length between 13 and 15 cm. The best crystal choice matching the requirements is provided by the new generation of Cerium doped Lutetium Yttrium Orthosilicate, LYSO, which has X_0 and R_M values (1.1 and 2 cm) comparable to the ones (0.9 and 2 cm) of the Lead Tungstanate, PbWO_4, with the advantage of a much larger light yield (× 300). On the negative side, LYSO shows a scintillation emission time (τ_{LYS O} = 40 ns) slower than PbWO_4 (τ_{PbWO} = 10 ns). However, from the basic scaling law of the timing resolution, σ_t = τ/ � Np.e., we expect LYSO to be a factor four more performant than PbWO_4.

In the final location of the CCALT inside KLOE-2, the presence of an axial magnetic field of 0.52 kGauss forces the usage of silicon based photodetectors. Due to the high photon yield, the readout with APDs is a valid solution since, at the lowest photon energy of 20 MeV, the collected photoelectrons will be around ∼ 10.000 which corresponds to ∼ 12 pC assuming an average gain of 300 and an amplification stage of ×25, well matching the ADC sensitivity of the KLOE calorimeter (100 fC/count). In the following, we specifically considered only the Hamamatsu S8664-55, which has an active area of 0.5×0.5 cm^2, fast timing characteristics and a quantum efficiency between 65 and 85% in the wavelenght range of interest (390∼ 500 nm) for the LYSO emission spectra.

In march 2009, we have built a medium size crystal matrix prototype with transversal radius larger than 2 R_m, longitudinal dimensions being constrained by budget limits to be between 13 and 15 cm (corresponding to 11 ÷ 12 X_0 of longitudinal containment). The prototype consists of an inner matrix of 10

Figure 3: Distribution of Q_{T OT} for single electron events at 100 MeV (top) with a logG fit superimposed and at 500 MeV (bottom) with a gaussian fit.

LYSO crystals readout by APD and an outer matrix, for leakage recovery, composed by 8 PbWO_4 cristals readout by standard Hamamatsu Bialcali photomultipliers of 1,1/8” diameter. To test the quality of the crystals offered by different vendors, the inner matrix has been assembled in three rows (see Fig. 2) composed (from bottom to top) as follows:

• 3 LFS crystals from Zecotek of 2×2×13 cm^3,

• 2 LYSO St.Gobain crystals of 1.5×1.5×15 cm^3, 1 LYSO St.Gobain + 1 LYSO Scionix crystals of 1.5×1.5×13 cm^3

• 1 LYSO St.Gobain crystal of 2×2×15 cm^3, 2 LYSO Scionix crystals of 2×2×13 cm^3.

The LFS from Zecotek is a Luthetium Fine Silicate crystal, with very similar properties to LYSO. Each crystal is wrapped with 100 μm of tyvek on the lateral faces, leaving free both the front and end faces, thus allowing to bring calibration light pulses through an external LED and a fast change of the photosensors readout. Each APD is inserted in a PVC mask with the amplifier soldered on its anode and then mecchanically positioned inside a stainless steel box closed by a PVC cap with only the electronic pins coming out for connection to HV and readout cables. An external holder takes the PMs in position for the readout of the outer crystals while allowing to press the boxes containing the APDs. The optical connection of the photosensors with the crystals is done with optical grease. The amplifiers are based on the MAR8A+ chip from Minicircuits, with a gain factor of 25 and a bandwidth of 1 GHz.

Figure 4: Shower example generated by a single event of 100 MeV electron in the matrix as shown by a full simulation based on Geant-4.

3. Test results with electron beams.

We have taken data at the Beam Test Facility, BTF, of LNF for two weeks in april 2009. The matrix was positioned at the center of the beam axis with an area delimited by a cross of two finger BC408 scintillators of 1×0.5×5 cm^3 dimensions, f 1, f 2. In most of the tests, the fingers were aligned in such a way to define a beam spot of 1 × 1 cm^2. In front of the fingers it was also present a beam position monitor, BPM, of the BTF group, consisting of sixteen horizontal and vertical scintillator strips readout by two Multi Anode PMs. Each strip is built by three 1 mm diameter scintillating fibers thus providing an accuracy below 1 mm on the beam localization.

We have triggered by using a replica of the spill signal from the Linac adjustable from remote in order to correctly put the signals in time. Prototype signals were splitted and discriminated by means of the standard KLOE calorimeter SDS boards. We acquire data with the KLOE-2 daq system, VME based, reading out KLOE ADC and TDC boards with a sensitivity of 100 fC/count and 50 ps/count respectively.

The BTF beam allows to set the average number of electrons, < Ne > arriving to the detector by adjusting dedicated collimators. However, we could not lower < Ne > below 1 for all beam energies and we cleaned the data offline by cutting on pulse height on the f1, f2 counters.

Observing the response of the prototype to single electrons, we realized that the outer matrix was not properly working due to an unexpected optical cross-talk between crystals. This was discovered by pulsing a UV LED (at 380 nm) in the front face of each LYSO crystal to let it scintillate and testing, both at the scope and with a fast data taking, the amount of signal observed on the contiguous crystals. We observe large cross-talk only on the PbWO_4 of the outer matrix. We believe this to be due by a cooperation between a light leak through the tyvek and the different amplification gains between PMs and APDs. In the following, we therefore report only results related to the inner matrix.

By using the UV LED, we have first equalized each channel at 10% level by proper HV adjusting. We have then calibrated the calorimeter response of each channel with minimum ionizing particles, m.i.p., crossing the calorimeter hortogonally to the crystal axis. To do this, we have collected cosmic ray runs with a dedicated trigger defined by two plastic scintillators po-

Figure 5: Dependence of the energy resolution on beam momentum

sitioned one above and one below the prototype. We get σ_{ped} of 5 counts and a m.i.p. peak, M_i, of around 100 counts for the smaller size crystals. The statistical precision on the peak determination is ∼ 1 %. The total response of the detector is then defined as:

Q_{T OT} = � (Q_i − P_i) × M_0/M_i, (1)

where Q_i and P_i are the collected charge and the pedestal of the i-th channel, M_0 represents an average calibration of all channels in counts and the calibration for the larger crystals is corrected for the different size. In Fig. 3, we show the distribution of Q_{tot} for a beam of 100 and 500 MeV respectively after having selected single electron events with a cut on the finger scintillators. A surviving fraction of events with more than 1 electron is still observed in the matrix, expecially at low energies. We have fit the distribution corresponding to one electron either with a simple gaussian, centered around the peak, or with a logarithmic gaussian, logG, function as follows:

N · exp(− 1 2σ^2_0 ln(1 − η σ_E

(E − E_{peak}))^2 − σ^2_0 2 ) (2)

where N is a normalization factore, η represents the asymmetry, E_{peak} the most probable value of the distribution, σ_0 = 2 2.36 sinh

−1(2.36η/2) and σ_E =^{FWHM}_{2.36} is the resolution. By performing a linear fit to the distribution of E_{peak} vs P_{beam}, we get a slope of 7.5± 0.1 counts/MeV which sets the M_i value to be ∼ 16 MeV for a small crystal consistently with an expected energy loss of ∼ 10 MeV/cm. At the running voltages of 410 V, the expected APD gain varies between 300-500 from which we estimate the light yield to be between 500 and 800 p.e./MeV.

To understand the different terms of the energy dependence of the energy resolution, we are carring out a full simulation of the prototype based on Geant-4. In Fig. 4, we report one example of 100 MeV electron shower as seen from the MC program. We have also simulated the dead space of 100 μm of tyvek between crystals and activated the optical transportation

Figure 6: Distribution of Q_{T OT} for events without electrons for runs with (up) 500 MeV, (down) 100 MeV beam.

of photons. Studies on photoelectron collection efficiency are still underways. This simulation indicates that there is a large leakage term between 5 and 4 % from 100 to 500 MeV.

In Fig. 5, we show the energy dependence of the energy resolution measured on data which has been fit with the following equation:

σ_E /E = a ⊕ b/(E/GeV) ⊕ c/ � E/GeV, (3)

where, accordingly to MC, we have fixed the constant term to be 5 %. We found b = 1.1% and c = 1.4% when using the gaussian fits to the spectra. If we repeat this procedure, for the fits with the logG function, we get b = 0.8% and c = 2.4%.

We have investigated the large b/E term by measuring the total detector noise with a gaussian fit to Q_{T OT} in events without any electron beam impinging (see Fig. 6). We find σ_Q = 4.2÷4.8 MeV which is slightly larger than the incoherent sum of σ_{ped} resulting to be 3.6÷3.8 MeV. A not negligible coherent noise is present and a much smaller (1/2) noise level has been previously measured in the electronic laboratory. However, the noise does not fully explain the large b term found. We are still investigating the origin of this contribution.

We have then determined the position resolution by comparing the reconstructed centroid done with the crystals with the position provided by the BPM of BTF. The centroids were defined as X_{pos} = � Q_iX_i/Q_{tot}. We observe a position resolution of 2.8 ÷ 3 mm at 500 MeV, which is slightly better than the crystal pitch as expected by the application of the centroid method.

We have finally reconstructed the calorimeter timing after correcting it, event by event, for the arrival time of the electrons in the LINAC spill. This was done by measuring the timing with the scintillators f1,f2. A very small pulse-height dependence on the calorimeter timing remains due to the usage of Constant Fraction Discriminators of KLOE. The weighted

Figure 7: Distribution of the average timing reconstructed with the inner matrix for 100 MeV electron beam.

energy average over all calorimeter, T_{clu}, was done after subtracting the average T_0 of each cell. In Fig. 7, the distribution of T_{clu} is shown for 100 MeV electron beam. A clean gaussian response is observed with a time resolution, σ_T , of ∼ 265 ps at 100 MeV. A similar study at other energies provided compatible results (e.g. σ_T ∼ 245 ps at 500 MeV). We have also measured the time jitter of the f1,f2 scintillators, by studying their time difference, which resulted to be of ∼ 240 ps thus explaining the week energy dependence of σ_T and pushing for a new measurement with a much reduced time-jitter.

Acknowledgments

The authors are indebt to many people for the succesfull realization of the matrix. In particular, we thank M. Lobello from Roma-3 university for the mechanical drawings, all the LNF mechanical shop for the realization of the support and APD boxes, expecially G. Bisogni, U. Martini and A. De Paolis. We also thank the BTF staff for providing the beam time and B. Buonomo for great support during the test beam data taking. The realization of the preamplifiers was done in collaboration with E.Reali from Roma-2 university.

References

[1] P.Raimondi, in: “Crab Waist Collisions in DAΦNE AND SUPER-B DE- SIGN”, Proceedings of EPAC08, Genoa, Italy (2008).

[2] F. Bossi et al, for the KLOE-2 collaboration,”A proposal for the Roll-In of the KLOE-2 detector”, LNF-Internal Note, 07/19 (2007).

[3] F. Bossi et al, for the KLOE collaboration, “Precision Kaon and Hadron Physics with KLOE”, Riv. Nuovo Cimento 031:531-623 (2009) [4] The KLOE collaboration, “Measurement of the BR(K_S → γγ) using a pure K_S beam with the KLOE detector”, JHEP 0805:051 (2008).