arXiv:1507.01408v3 [physics.med-ph] 10 Aug 2015

3D super-resolved in vitro multiphoton microscopy by saturation of excitation

Anh Dung Nguyen,^{1, ∗} Franois Duport,^1 Arno Bouwens,^1 Frédérique Vanholsbeeck,^2 Dominique Egrise,^3 Gaetan Van Simayes,^3 Serge Goldman,^3 Philippe Emplit,^1 and Simon-Pierre Gorza^1 1Université libre de Bruxelles, Service OPERA Photonique, Avenue F.D. Roosevelt 50 CP194/5, 1050 Brussels, Belgium 2University of Auckland, The Dodds-Walls Centre for Photonic and Quantum Technologies & Department of Physics, Private bag 92019, Auckland 1142, New Zealand 3Université libre de Bruxelles, Department of Nuclear Medecine, ULB-Hopital Erasme, 808 Route de Lennik, 1070 Brussels, Belgium

We demonstrate a significant resolution enhancement beyond the conventional limit in multiphoton microscopy (MPM) using saturated excitation of fluorescence. Our technique achieves superresolved imaging by temporally modulating the excitation laser-intensity and demodulating the higher harmonics from the saturated fluorescence signal. The improvement of the lateral and axial resolutions is measured on a sample of fluorescent microspheres. While the third harmonic already provides an enhanced resolution, we show that a further improvement can be obtained with an appropriate linear combination of the demodulated harmonics. Finally, we present in vitro imaging of fluorescent microspheres incorporated in HeLa cells to show that this technique performs well in biological samples.

INTRODUCTION

Since its introduction in 1990 [1], multiphoton microscopy (MPM) rapidly evolved to become the workhorse for 3D fluorescence imaging in many laboratories. MPM is arguably one of the most influential recent inventions in optical microscopy because it offers a combination of minimally invasive imaging with high resolution, thin optical sectioning and deep penetration in highly scattering samples such as biological tissues [2]. These properties all stem from the process of multiphoton fluorescence excitation. In order to achieve efficient multiphoton excitation, photons need to be concentrated both in space by focussing the excitation beam, and in time by using pulsed laser sources. Fluorescence is therefore only excited in a narrow volume. Moreover, because the excitation spectra of common fluorescent molecules are in the visible spectrum, the wavelengths used for MPM are in the near-infrared such that optical sample damage is much reduced [3, 4].

A second important development in optical microscopy has been the development of super-resolution microscopy which achieves resolutions beyond the diffraction limit. In essence, super-resolved imaging requires the sample to respond non-linearly to the illumination light. Such a non-linearity can be achieved by, among other techniques, saturation of excitation [5] and stimulation emission depletion (STED) [6]. Bringing super-resolution to MPM has the potential to enable studying biological samples at the depths offered by MPM with a resolution beyond the diffraction limit.

STED has been applied successfully for two-photon microscopy (2PM), but only to improve the lateral resolution [7]. Moreover, STED requires the use of a highintensity visible wavelength depletion beam, increasing the invasiveness of the method. Although STED microscopy can also improve the axial resolution beyond the diffraction limit [8], it requires a carefully aligned and relatively complex optical system, often including an additional depletion laser source.

In this work, we present a simple approach to achieve 3D super-resolution in MPM, based on saturation of fluorescence excitation [5]. This method can easily be implemented on an existing multiphoton microscope, as it only requires (a) temporal modulation of the excitation beam using an off-the-shelf intensity modulator, and (b) demodulation of the detected fluorescence signal with an electronic lock-in amplifier. Hence, no additional laser sources or beam shaping optics are required, and only near-infrared light is incident on the sample.

Images of fluorescent microspheres on a coverslip prove that this technique can be implemented successfully in MPM and improves not only the lateral but also the axial resolution. Further experiments with microspheres integrated in HeLa cells produced encouraging results demonstrating the feasibility of this method for in vitro studies.

PRINCIPLE OF THE METHOD

The technique proposed in this work is based on the saturation of the probability of excitation of the fluorophore in MPM. Until recently, saturation has been considered harmful for imaging and has been avoided [9, 10]. It is possible, however, to show that in a saturation regime, the relation between the intensity of excitation and the emitted fluorescence intensity is no longer linear and contains nonlinearities that can be used to improve the lateral as well as the axial resolution [5, 11, 12].

In a two-photon excitation process, assuming a delay between pulses much longer than the fluorescence lifetime, the probability of exciting a fluorophore at a given position with one excitation laser pulse is given by P_{exc} = 1 − e^{−ασ2P A P 2P SF 2 (x,y,z)}, (1) where α is a coefficient which depends on the pulse’s shape and duration, and on the repetition rate of the laser, σ_{2P A} is the two-photon absorption cross-section of the fluorophore, P the average power of the excitation beam and P SF (x, y, z) is the illumination point spread function. To extract the nonlinearities that appear when saturation occurs, a temporal modulation of frequency ν_m is applied to the excitation beam [5]. The probability of excitation can then be written P_{exc} = 1 − e^{−ασ2P AP 2[1+cos(ωt)]2P SF 2 (x,y,z)}, (2) where ω = 2πν_m.

At high excitation power, higher order harmonics in the fluorescence signal will become non-negligible, as illustrated in Fig. 1(a).

By developing the 1 − e^{−x} term up to the 4^{th} order in Eq. (2), and rearranging the expression, the power in the harmonics can be calculated P^{signal} ν_m : 2P SF 2 −^7_2 P SF^4 +^{33}_8 P SF^6, P^{signal} 2ν_m : 1 2 P SF^2 −^7_4 P SF^4 +^{495}_{192} P SF^6, P^{signal} 3ν_m : 1 2 P SF^4 −^{55}_{48} P SF^6. (3)

These equations show that the harmonics in the detected fluorescence signal contain terms with the fourth and sixth powers of the illumination PSF. In standard two-photon microscopy, the detected signal contains P SF^2. Hence, the resolution is improved by at least a factor of √ 2 when using the third harmonic. Figure 1(b) shows the dependence of the harmonics’ strength S_{emission} on the illumination intensity.

FIG. 1. Apparition of higher harmonics in the saturation regime. (a) Spectrum of the signal of the detected fluorescence at a modulation frequency of νm = 5 kHz. (b) Power in the n-th harmonic of the detected fluorescence signal (Semission) for increasing power on the sample.

By making the appropriate linear combination of the harmonics in Eq. (3), it is possible to obtain a signal which is dependent only on the sixth and higher powers of the P SF : Linear combination signal = P^{signal}_{νm} −4P^{signal}_{2νm} −7P^{signal}_{3νm} . (4) The linear combination signal (∝ P SF^6) therefore achieves an additional resolution improvement by a factor of √ 3/2 over the third harmonic signal P^{signal}_{3νm} assuming a gaussian illumination profile.

MATERIAL AND METHODS

Experimental setup

FIG. 2. The experimental setup is based on a conventional multi-photon microscope with an acousto-optic modulator to modulate the fluorescence excitation intensity, and a lock-in amplifier to extract the harmonics of the fluorescence signal detected by the PMT.

The experimental setup, shown in Fig. 2, is built around a commercial two-photon microscope (Model FV1000-MPE, based on a BX61WI upright microscope body, Olympus Corp.). The configuration of the microscope is the U-MWG2 which has two detection channels with filters allowing the detection of fluorescence from 510 to 550 nm and from 600 to 690 nm. The microscope objective is the Olympus XLPN25XWMP (numerical aperture 1.05, magnification 25×). The multiphoton excitation laser system used in our experiments is a Spectra-Physics Ti:Sa Tsunami pumped by a Millenia ProS-Series. This laser is tunable over a wavelength range of 840 nm to 980 nm. It has a repetition rate of 82 MHz and is capable of producing 150 fs pulses at a wavelength of 880 nm. The excitation light beam is temporally modulated by an acousto-optic modulator (MT80-A1-IR, AA Opto Electronic Sa.). A lock-in amplifier (HF2LI, Zurich Instruments AG) demodulates the signal from the photomultiplier tubes (R3896, Hamamatsu Photonics K.K.) of the microscope at the first, second and third harmonics. The linear combination of harmonic signals is generated by an analog summation circuit such that no offline processing is required to generate the final image. The microspheres used for the different experiments are Dragon Green fluorescent microspheres (Bangs Laboratories, (480, 520)).

Sample preparation

Stock cultures of Hela cells were maintained in RPMI (Roswell Park Memorial Institute medium) supplemented with 10% (v/v) deactivated fetal bovine serum, 2 mM L-glutamine and 50 μg/ml gentamycin (complete medium, Lonza) at 37^∘C in a humidified atmosphere of 95% air/5% CO_2. Cells were recovered by trypsinisation (trypsin-EDTA, Lonza).

Fluorescent microspheres (Bangs Laboratories) were used, with a mean diameter of 0.51 μm (FC03F). The suspensions of beads were homogenized (rotation) overnight at 4^∘C before use. 5 × 10^5 cells were mixed with 10 μl of bead suspension and 4 ml complete culture medium, plated in 10 cm^2 culture wells and incubated for 24 hours at 37^∘C in a humidified atmosphere of 95% air/5% CO_2. At the end of the incubation the cells were recovered by trypsinisation. To eliminate the beads not adsorbed by the cells, cells were diluted in 2 ml medium, put in a 3 μm pore size cell culture insert (Greiner Bio-one) and briefly centrifuged at 200 g. Cells were recovered by washing the insert with culture medium, plated on glass coverslips at a density of 5 × 10^4 cells/cm^2, and incubated overnight at 37^∘C under an atmosphere of 5% CO_2. The coverslips were then rinsed 3 times with phosphate buffered saline (PBS), fixed with 4% Paraformaldehyde (PFA) during 10 minutes and rinsed again with PBS.

Acquisition parameters

All imaging experiments were performed at a modulation frequency ν_m of 17 kHz, and the first, second (34 kHz) and third (51 kHz) harmonics were demodulated. The images containing only the first and second harmonics were obtained with 10 mW excitation power on the sample. To obtain the images of the third harmonic as well as the linear combination, the excitation power was increased to 25 mW to produce a sufficiently strong signal. The complete 2D field of view (x, y) of 508.68×508.68 μm^2 is sampled at 4096×4096 pixels with a pixel dwell time of 200 μs, which corresponds to 3.4 modulation periods at 17 kHz, 6.8 at 34 kHz and 10.2 at 51 kHz. (x, y, z) scans are stacks where each (x, y)-slice (same field of view as for 2D images) is sampled at 2048 × 2048 pixels.

RESULTS

Microspheres on a coverslip

To demonstrate the resolution improvement of our method, 0.19 μm microspheres were imaged. First we determined the power needed to obtain saturation by measuring the harmonics in the fluorescence signal. To this end, we analyzed the signal from the PMT with an electronic spectrum analyzer (ESA) (8564E, Hewlett Packard), and increased the illumination power until the harmonics rose above the noise floor of the ESA as shown in Fig. 3(a). Note that, due to the narrowband detection of the lock-in amplifier, the actual noise floor of the imaging system is expected to be lower than the noise floor measured by the ESA (−70 dBm).

FIG. 3. Harmonics in the detected fluorescence signal from 0.19μm microspheres on a coverslip. (a) Frequency spectra of the fluorescence signal emitted by the microspheres for increasing average power. (b) Spectral intensities of the three first harmonics in the fluorescence signal. Linear fits of the data are shown as black lines. The particle concentration was 10 mg/mL and the frequency of the excitation modulation νm was 17 kHz.

The third harmonic clearly emerged from the noise for an illumination power of approximately 30 mW on the sample. At this illumination power, the signal-to-noise ratio was measured to be 15 dB and was sufficient to acquire an image. The demodulated fluorescence intensity at a frequency of ν_m (17 kHz), 2ν_m (34 kHz) and 3ν_m (51 kHz) is shown in Fig. 3(b). As expected from Eq. (3) and Fig. 1(b), the curves of the fundamental and second harmonics are linear and have the same slope, while the slope of the third harmonic curve is twice as steep.

For the fundamental (17 kHz) and the second harmonic (34 kHz) [Fig. 4(a) and (b)], the spatial resolution in x and y was measured to be the same as in conventional MPM. The images obtained using the third harmonic [Fig. 4(c)] and the linear combination of the three first harmonics given by Eq. (4) [Fig. 4(d)] clearly show a resolution improvement. The intensity profiles shown in Fig. 4(e) confirm this improvement: for the first and the second harmonics, the two microspheres are not resolved, they can just be resolved by the third harmonic. The linear combination resolves the two microspheres even better.

The full-width at half-maximum (FWHM) was measured on the leftmost microsphere to be approximately 470 nm for the two first harmonics, 340 nm for the third and 285 nm for the linear combination signal. These measurements fit with the estimated theoretical improvement of the resolution.

The same analysis was performed in the axial direction to demonstrate the improvement of the axial res-

FIG. 4. Characterization of the lateral resolution improvement using 0.19 μm microspheres on a coverslip. (a-d) Images in the (x, y) plane reconstructed with the first, second, and third harmonics and the linear combination of these three harmonics. The (x, y)-scan size is 48.36 × 56.54 μm^2 (390 × 456 pixels) resulting in a total acquisition time of ∼ 35 s. (e) Signal profiles along the orange line in (a-d). The background intensity was measured in an empty area and subtracted everywhere.

olution. Figure 5 shows the four different (x, z) planes obtained with the first, second, and third harmonics and the linear combination. The results confirm that the axial resolution is also improved using saturation of the excitation. The axial FWHM was measured to be approximately 3.4 μm for the two first harmonics, 2.1 μm for the third and 1.8 μm for the linear combination signal. Again, these values match the estimated theoretical resolution improvement.

Microspheres in HeLa cells

To demonstrate the applicability of this multiphoton super-resolution technique to biological samples, we went on to image 0.51 μm microspheres incorporated in HeLa cells. Figure 6 shows (y, z) sections of 3D images constructed from the first, second, and third harmonics and the linear combination of those three harmonics. 3D renderings were generated with the Imaris software (Bitplane AG).

As for the microspheres on a coverslip, an improvement of the resolution in all three dimensions is observed when imaging at the third harmonic and a further improvement is obtained by the linear combination of the three first harmonics.

FIG. 5. Characterization of the axial resolution improvement using 0.19 μm microspheres on a coverslip. (a-d) Images in the (y, z) plane reconstructed with the first, second, and third harmonics and the linear combination of these three harmonics. The (y, z)-sections are taken from (x, y, z)-stacks consisting of 19 (x, y)-slices of 49.10 × 58.53 μm^2 (192 × 236 pixels), resulting in a total acquisition time of 3 minutes per 3D stack. (e) Signal profiles along the yellow line in (a-d). The background intensity was measured in an empty area and subtracted everywhere.

FIG. 6. Microspheres of 0.51 μm in HeLa cells. (a-d) Images in the (y, z) plane show a clear improvement of the resolution in the lateral and axial dimensions. (e-h) 3D renderings for the first, second and third harmonics and the linear combination of the three first harmonics. The (x, y, z)-stacks consisted of 17 (x, y)-slices of 169.6 × 169.6 μm^2 (800 × 800 pixels), resulting in a total acquisition time of 36 minutes per 3D stack.

DISCUSSION AND CONCLUSIONS

In this work, we demonstrated experimentally that saturation of fluorescence excitation can be exploited to obtain a 3D resolution improvement in MPM. Implementation of this technique on an existing MPM system is straightforward since it only requires modulating the excitation beam and demodulating the detected fluorescence signal. The method is also cost-effective because it does not require additional laser sources or microscope objectives to improve the 3D resolution. Moreover, the technique does not employ delicate 3D beam-shaping optics, which are sensitive to misalignment over time.

Images of fluorescent microspheres on a coverslip and integrated in HeLa cells show that resolution is indeed improved in all three spatial dimensions by a factor of

√ 2 when imaging at the demodulated third harmonic of the detected fluorescence signal. Furthermore, we showed that the 3D resolution can be improved by an additional factor of √ 3/2 when the appropriate linear combination is made of the first three harmonics. This linear combination of harmonics is made by an analog circuit such that no subsequent signal processing is required. For comparison, our super-resolution MPM obtains the same lateral resolution as a 400 nm illuminated linear confocal microscope (assuming the same beam characteristics and objective), while conserving the advantages of classical MPM (i.e. near-infrared illumination and increased depth of penetration).

To obtain the required saturation regime and get a sufficiently high signal-to-noise ratio for the harmonic signals, an average illumination power of 25 mW at 880 nm was delivered on the sample. Interestingly, this rather modest illumination power suffices to obtain a near twofold 3D resolution enhancement in MPM. Moreover, since the sample is illuminated only at near-infrared wavelengths, our approach is likely to be less invasive, in terms of photo-toxicity, than the super-resolution techniques based on STED.

With the present acquisition parameters, relatively few modulation periods are measured at every image voxel. Even though the resulting signal-to-noise ratio was sufficient for imaging, there is room for improvement by increasing the number of periods of modulation of the excitation signal per pixel. This can be done by increasing the pixel dwell time (if the acquisition time is not critical) or by using a higher modulation frequency (but keeping a sufficient number of pulses per modulation period). An improved signal-to-noise ratio could allow using a lower illumination power while still obtaining a sufficiently strong signal to reconstruct the image. This could be particularly interesting to investigate for future work, especially for in vivo applications where the power damage threshold is critical.

Theoretically, the resolution improvement is unlimited since higher harmonics than the third one can be used. However, as for all super-resolution methods based on saturation of the excitation or STED, the attainable resolution is ultimately limited by the damage threshold of the sample. At high illumination powers, photobleaching and phototoxicity become critical in biological samples. This problem can be reduced using fluorophores with high two-photon absorption cross section or quantum dots, provided that their fluorescence lifetime is sufficiently short compared to the modulation period.

ACKNOWLEDGMENTS

This work was supported by the Institut Interuniversitaire des Sciences Nuclaires (IISN), Grant No 4.4510.09 and by the Belgian Science Policy Office (BELSPO) Interuniversity Attraction Pole (IAP), Project IAP 7/35 Photonics@be.

Frédérique Vanholsbeeck acknowledges the support of a University of Auckland grant for research and study leave, a “Brains (Back) to Brussels” grant (B2B 2012-2- 08) from the Brussels institute for research and innovation (Innoviris) and a “Missions Scientifiques Professeurs Etrangers” grant from F.N.R.S. that supported her stay at the Universite libre de Bruxelles.

∗ anhdung@ulb.ac.be [1] W. Denk, J.H. Strickler, and W.W. Webb, “Twophoton laser scanning fluorescence microscopy,” Science. 248(4951), 73-76 (1990).

[2] J. Pawley and B.R. Masters, “Handbook of confocal microscopy,” Optical Engineering, International Society for Optics and Photonics, (2006).

[3] W. Denk and K. Svoboda, “Photon upmanship: why multiphoton imaging is more than a gimmick,” Neuron. 18(3), 351-357 (1997).

[4] W.R. Zipfel, R.M. Wiliams and W.W. Webb, “Nonlinear magic: multiphoton microscopy in the biosciences,” Nature Biotech. 21(11), 1369-1377 (2003).

[5] K. Fujita, M. Kobayashi, S. Kawano, M. Yamanaka and S. Kawata, “High-Resolution Confocal Microscopy by Saturated Excitation of Fluorescence,” Phys. Rev. Lett. 99(22), 228105 (2007).

[6] S.W. Hell and J. Wichmann, “Breaking the diffraction resolution limit by stimulated emission: stimulatedemission-depletion fluorescence microscopy,” Opt. Let. 19(11), 780-782 (1994).

[7] G. Moneron and S.W. Hell, “Two-photon excitation STED microscopy,” Opt. Exp. 17(17), 14567-14573 (2009).

[8] D. Wildanger, R. Medda, L. Kastrup and S.W. Hell, “A compact STED microscope providing 3D nanoscale resolution,” Journal of Microsopy. 236(1), 35-43 (2009).

[9] G.C.Cianci, J. Wu and K.M. Berland, “Saturation modified point spread function in two-photon microscopy,” Microsc. Res. Tech. 64, 135-141 (2004).

[10] K. Visscher, G.J. Brakenhoff and T.D. Visser, “Fluorescence saturation in confocal microscopy,” Journal of Microsopy. 175(2), 162-165 (1994).

[11] M. Yamanaka, S. Kawano, K. Fujita, N.I. Smith and S.

Kawata, “Beyond the Diffraction-limit: Biological Imaging by Saturated Excitation Microscopy,” J. Biomed. Opt. 13(5), 050507 (2008).

[12] M. Yamanaka, Y. Yonemaru, S. Kawano, K. Uegaki, N.I. Smith, S. Kawata, and K. Fujita, “Saturated excitation microscopy for sub-diffraction-limited imaging of cell clusters,” J. Biomed. Opt. 1(12), 126002 (2013).