arXiv:1611.01596v1 [astro-ph.IM] 5 Nov 2016

MNRAS 000, 1–7 (2015) Preprint 30 May 2019 Compiled using MNRAS L^ATEX style file v3.0

A high precision polarimeter for small telescopes

Jeremy Bailey,^{1,2⋆} Daniel V. Cotton,^{1,2} Lucyna Kedziora-Chudczer^{1,2}

1School of Physics, UNSW Australia, NSW 2052, Australia 2Australian Centre for Astrobiology, UNSW Australia, NSW 2052, Australia

Accepted XXX. Received YYY; in original form ZZZ

ABSTRACT We describe Mini-HIPPI (Miniature HIgh Precision Polarimetric Instrument), a stellar polarimeter weighing just 650 grams but capable of measuring linear polarization to ∼ 10^{−5}. Mini-HIPPI is based on the use of a Ferroelectric Liquid Crystal (FLC) modulator. It can easily be mounted on a small telescope and allows study of the polarization of bright stars at levels of precision which are hitherto largely unexplored. We present results obtained with Mini-HIPPI on a 35 cm telescope. Measurements of polarized standard stars are in good agreement with predicted values. Meaurements of a number of bright stars agree well with those from other high-sensitivity polarimeters. Observations of the binary system Spica show polarization variability around the orbital cycle.

Key words: polarization – instrumentation: polarimeters – techniques: polarimetric

1 INTRODUCTION

Recently a number of polarimeters have been developed that can measure stellar polarization at parts-per-million levels (Hough et al. 2006; Wiktorowicz & Matthews 2008; Wiktorowicz & Nofi 2015; Bailey et al. 2015). All of these instruments are on 3–5m class telescopes and one of their main aims has been the measurement of polarized reflected light from extrasolar planets (e.g. Lucas et al. 2009; Wiktorowicz et al. 2015; Bott et al. 2016). However, such instruments can also provide valuable data on the polarization of bright stars. For example, surveys of bright stars (Bailey et al. 2010; Cotton et al. 2016a) have identified interstellar polarization in relatively nearby stars, and found that intrinsic polarization in normal stars is more common than had previously been thought. For studies of the polarization of bright stars it is not necessary to use a telescope as large as 3–5m.

Most of these instruments have been based on the photoelastic modulator (PEM) technology originally developed by James Kemp (Kemp 1969; Kemp & Barbour 1981; Kemp et al. 1987). However, Bailey et al. (2015) showed that high polarimetric sensitivity could also be achieved using Ferroelectric Liquid Crystal (FLC) modulators. The compact size and simple drive requirements of FLC’s make them ideally suited to the development of extremely small and lightweight instuments. Here we describe a FLCbased polarimeter sufficiently compact and lightweight to be mounted on even the smallest telescopes.

⋆ E-mail: j.bailey@unsw.edu.au

2 INSTRUMENT DESCRIPTION

2.1 Overview

Mini-HIPPI is based on the same design principles as the HIPPI instrument (Bailey et al. 2015) used on the 3.9m Anglo-Australian Telescope (AAT). HIPPI was commissioned in 2014 and has been used regularly on the AAT for a number of applications (Bailey et al. 2015; Cotton et al. 2016a; Bott et al. 2016; Marshall et al. 2016). Mini-HIPPI shares the same modulators, detectors, and data acquisition and data reduction software.

The optical system of Mini-HIPPI is shown in figure 1 and a picture of the instrument is given in figure 2.

In order of their position on the light path from the telescope the components of the instrument are as follows:

• A Thorlabs PRM1Z8 motorized rotation stage allows the whole instrument to be rotated.

• The first optical element is the Ferroelectric Liquid Crystal (FLC) modulator. This is an LV1300-AR-OEM device from Micron Technology. It is designed for the 400-700 nm wavelength range. It operates as an electrically switched half-wave plate driven by a ±5 V square wave which causes the optical axis to switch between two orientations 45 degrees apart. The normal modulation frequency used is 500 Hz. Further details on these devices and the method of driving them can be found in Bailey et al. (2015). The FLC is a temperature sensitive device and is therefore mounted in a temperature controlled lens tube (Thorlabs SM1L10H) and maintained at a temperature of 25^∘C.

• Behind the lens tube carrying the FLC, is a rotatable

© 2015 The Authors

2 Bailey et al.

Figure 1. Optical system of Mini-HIPPI

back-end module that carries the aperture, polarizing prism and detectors. This can be adjusted in rotation to minimize the instrumental polarization as described below. The aperture is mounted on a removable slide to allow change of aperture sizes.

• The polarizing prism is a calcite Glan-Taylor prism (Thorlabs GT5-A) of 5mm aperture. This splits the light into two beams with orthogonal polarizations, one passing straight through the prism, and one directed upwards at an angle of 68 degrees to the incident beam.

• The two beams are fed to two Hamamatsu H10720-210 Photomultiplier Tube (PMT) modules with attached transimpedance amplifiers, identical to those used with HIPPI (Bailey et al. 2015).

Apart from the commercial components described above, the construction of the instrument is entirely by 3D printing in black ABS plastic.

The main differences from HIPPI are:

• The use of a Glan-Taylor prism rather than a Wollaston prism. This makes possible a much shorter light path helping to keep the instrument compact.

• The lack of any lenses. HIPPI used lenses to collimate the beam through the prism, and Fabry lenses to image the pupil onto the detectors. The extremely short light path in Mini-HIPPI allows the diverging beam from the aperture to be fed straight to the detectors while still being well within the 8 mm diameter of the photocathodes.

• The lack of a motorized back-end rotation. This means that Mini-HIPPI only has two stages of modulation (The primary FLC 500 Hz modulation, and the whole instrument rotation) whereas HIPPI has three.

• Mini-HIPPI does not have a filter wheel. It has been designed to allow a fixed 25 mm diameter filter to be fitted, but we normally use Mini-HIPPI without a filter, giving a broad bandpass defined mostly by the PMT response. This helps to maximize the signal with small telescopes.

While all the observations described here use bluesensitive PMTs and a modulator optimized for blue wavelengths, it is realatively straightforward to adapt the instrument for red wavelengths by fitting a red-optimized FLC modulator and red-sensitive Hamamatsu H10720-20 PMT modules with extended red multialkali photocathodes sensitive out to 920 nm.

Figure 2. Mini-HIPPI with the main components indicated

Figure 3. Mini-HIPPI data acquistion and control system

2.2 Data Acquisition System

The data acquisition system for mini-HIPPI is similar to that for HIPPI and uses essentially the same software developed in the National Instruments LabVIEW graphical programming environment. However, the control electronics has been made more compact by using an Intel NUC miniature PC, running Windows 7 Professional, and connected to two National Instruments USB-6211 Data Acquisition modules. These provide analog inputs to read the signals from the PMTs and analog outputs to provide the drive waveform to the FLC modulator as well as the control signals to set the PMT gain and HT voltage.

In operation the signals are read at a 10 microsecond sample time, synchronized with the modulation using a trigger pulse derived from the FLC drive waveform. The signals are folded over the modulation period (2 milliseconds for our standard 500 Hz modulation frequency) and written to output files after an integration time of typically one second. The format of the output files is identical to those written by HIPPI and can be analysed using the same data reduction software.

A simple quick-look data reduction system built into the acquisition software allows polarization levels to be assessed in real time.

MNRAS 000, 1–7 (2015)

High Precision Polarimeter 3

2.3 Observing Procedure

As explained by Bailey et al. (2015) polarimeters of this design inherently have a substantial instrumental polarization due to multiple reflection effects in the FLC modulator (Gisler et al. 2003; de Juan Ovelar et al. 2012). The key to achieving high precision is designing the instrument in a way that this instumental polarization can be removed. This is possible because the instrument only measures one Stokes parameter at a time. By rotating the back-end of the instrument relative to the FLC, a one-time adjustment when the instrument is set up, a setting can be found at which the instrumental polarization is orthogonal to the Stokes parameter being measured. This reduces the instrumental polarization to around 100 parts-per-million or less.

This residual instrumental polarization is then eliminated by repeating all observations with the whole instrument rotated through 90 degrees relative to the telescope and sky. This reverses the measured sign of any real polarization, while leaving the instrumental polarization unchanged. In practice we observe at four position angles (0, 45, 90, 135 degrees). One Stokes parameter is obtained from the 0 and 90 degree observations, and the other from the 45 and 135 observations.

This measured polarizations will still include any polarization introduced by the telescope (Telescope Polarization or TP). This must be corrected by observations of low polarization stars as described later.

2.4 Data Reduction

The data reduction software used for Mini-HIPPI is identical to that for HIPPI described in Bailey et al. (2015). The polarization is determined by fitting a modelled curve to the observed modulation waveforms using a Mueller matrix model of the instrument. The Mueller matrix for the instrument is the matrix M that transforms the input Stokes vector of the source being observed s_{in} with components (I, Q, U, V) into the output Stokes vector sout seen at the detector.

sout = Ms_{in} (1)

The matrix M is itself the product of the Mueller matrices for each of the optical elements in the system as follows:

M = M_{Pol}MRetM_{Depol} (2)

where M_{Pol} describes the polarizing effect of the Wollaston prism, MRet describes the retardance of the modulator, and M_{Depol} describes the depolarizing effect of the modulator. The matrix M varies around the modulation cycle because the retardance and depolarization vary. We can determine the form of this variation by a laboratory calibration procedure where known polarization states are input to the instrument using a lamp and rotatable polarizer. From the set of matrices M at different modulation phases we can then define a system matrix W which when multiplied by the input Stokes vector gives the observed vector of output intensities at each modulation phase. From the inverse of this matrix (Sabatke et al. 2000) we can then derive the input polarization from the observed modulation curve. Full details of this procedure are given in Bailey et al. (2015).

Table 1. Effective wavelength and modulation efficiency for different spectral types according to bandpass model

Spectral Effective Modulation Type Wavelength (nm) Efficiency (%)

B0 V 438.7 69.2 A0 V 458.2 78.8 F0 V 467.5 80.4 G0 V 478.2 82.2 K0 V 490.3 85.0 M0 V 509.9 86.6 M5 V 510.4 86.2

Figure 4. The black line is the instrument response of Mini- HIPPI obtained by combining the atmospheric and instrument throughput and the response of the PMT detector. The blue and red curves show the effective response when observing an A0 star and an M0 star (using Kurucz model spectra). The dashed lines show the effective wavelength in each case which shifts from 458.2 nm for the A0 to 509.9 nm for the M0 star. See also table 1.

2.5 Bandpass Model

Mini-HIPPI is normally used without a filter and therefore has a broad spectral response from around 350 – 700 nm but with the peak in the blue at around 440 nm due to the response of the Ultra-bialkali photocathodes used. This broad response means that the precise effective wavelength depends on the colour of the object being observed. The modulation efficiency of the polarimeter is also wavelength dependent. The FLC modulators used are designed to be half-wave plates at about 500 nm. The modulation efficiency falls off at wavelengths either side of this.

To account for these effects we use a bandpass model as described in Bailey et al. (2015). This model allows us to specify a source spectrum, usually one of the stellar atmosphere models from Castelli & Kurucz (2004), optionally correct it for interstellar extinction, and then apply atmospheric transmission and instrument response functions to

MNRAS 000, 1–7 (2015)

4 Bailey et al.

Figure 5. Mini-HIPPI mounted on the 35cm telescope at UNSW Observatory

allow determination of the effective wavelength, and bandpass averaged modulation efficiency. The typical results for nearby stars (no interstellar extinction) are given in table 1. Examples of response curves are given in figure 4.

3 OBSERVATIONS

The observations described here were used to test and evaluate Mini-HIPPI and were made at the UNSW Observatory located at the Kesington campus in Sydney. The telescope is a 35 cm Schmidt Cassegrain (Celestron-14) mounted on a Losmandy German Equatorial mount. We attached Mini- HIPPI to the telescope via an acquisition module that allowed the field-of-view to be imaged via a flip-in mirror, that can then be removed to pass the starlight through to the instrument. The weight of the instrument plus acquisition module is 1.2 kg. It attaches to the telescope through a standard 2-inch eyepiece fitting. The instrument mounted on the telescope is shown in figure 5. A camera mounted on an attached 80 mm guide telescope is used for field acquisition and autoguiding.

The focal length of the telescope is 3900 mm and our 1.2 mm aperture corresponds to 63 arc seconds on the sky. With this large aperture and our city site, which is never really dark even when the Moon is absent, it is important to correct for sky background. Each observation consists of measurements at four rotator position angles, and at each of these angles we also make at least one sky observation. For long integrations on fainter targets we take sky measurements before and after the target observations at each angle. The sky modulation curve is subtracted from the target observation before further analysis.

3.1 Telescope Polarization

All observations need to be corrected for telescope polarization (TP). The TP is determined by observations of nearby stars that we believe to have very low polarization. The low polarization calibrators were all stars we have used previously for this purpose with HIPPI (Bailey et al.

Table 2. Low polarization star measurements to determine telescope polarization

Block Star Date P (ppm) θ (deg)

May 2016 Sirius May 13 95.2 ± 7.8 93.2 ± 2.3 Sirius May 14 98.8 ± 8.3 91.9 ± 2.5 Sirius May 17 99.8 ± 8.2 90.6 ± 2.4 Sirius May 19 75.8 ± 8.7 83.0 ± 3.2 Average 91.6 ± 4.1 90.1 ± 1.3

Jun 2016 Sirius Jun 9 62.1 ± 11.2 93.6 ± 5.5 β Leo Jun 9 67.5 ± 19.7 89.2 ± 8.4 Sirius Jun 10 54.2 ± 6.6 93.3 ± 3.6 Sirius Jun 15 61.3 ± 12.5 91.9 ± 6.0 Sirius Jun 16 57.2 ± 7.6 91.2 ± 3.7 Average 60.4 ± 5.6 91.8 ± 2.7

Jul-Sep β Leo Jun 29 31.7 ± 18.3 88.1 ± 16.2 2016 β Leo Jun 29 83.8 ± 18.4 82.0 ± 6.4 β Vir Jun 29 116.9 ± 29.5 95.6 ± 7.3 β Leo Jul 1 64.4 ± 19.3 95.3 ± 8.2 β Leo Jul 1 59.8 ± 19.7 93.6 ± 9.6 β Hyi Jul 23 50.5 ± 26.7 93.0 ± 14.7 β Hyi Jul 23 89.4 ± 29.0 81.4 ± 19.3 β Leo Jul 24 101.1 ± 23.3 95.2 ± 6.5 β Leo Jul 25 32.9 ± 22.0 88.9 ± 19.0 α Ser Aug 11 89.4 ± 21.5 106.1 ± 6.8 β Hyi Aug 15 97.1 ± 19.3 108.5 ± 5.7 β Hyi Aug 31 77.8 ± 19.0 97.5 ± 7.0 Sirius Sep 19 59.3 ± 7.8 86.4 ± 4.1 Sirius Sep 19 83.0 ± 8.4 86.3 ± 3.2 Average 71.0 ± 5.6 93.5 ± 2.3

2015; Cotton et al. 2016a) or PlanetPol (Hough et al. 2006; Bailey et al. 2010). They are Sirius, β Leo (BS 4534), β Vir (BS 4540), β Hyi (HIP 2021, BS 98) and α Ser (BS 5854). Based on our previous observations of these objects we believe they all have polarizations of ∼ 5 ppm (parts per million) or smaller.

We determined the TP separately for our blocks of observations in May 2016, Jun 2016 and Jul-Sep 2016. The instrument was removed from the telescope and remounted and adjusted between each of these blocks. The TP adopted was the mean of the polarization of all calibrators observed within each block, and this was used to correct all other observations within the block.

When Sirius was observable this was our preferred calibrator as it could be observed to a precision of usually better than 10 ppm in a short observation. The other objects were fainter and typical precisions of individual observatons were ∼ 20 – 30 ppm.

Between the May and June observing blocks the instrument was modified by fitting with an improved aperture plate that removed some light leaks, and was remounted with a different orientation on the telescope. This probably explains the change in TP between these runs.

3.2 High Polarization Stars

We have observed a number of high polarization stars. These are used to calibrate the position angle of polarization and also provide a check on the efficiency calibration of the in-

MNRAS 000, 1–7 (2015)

High Precision Polarimeter 5

Table 3. Predicted and observed polarization for polarized standard stars

Star Spectral Type E_{B−V} R_V P_{max} λ_{max} K θ (deg) P (%) θ (deg) Refs Predicted Observed Observed

HD 84810 F8-K0Ib V 0.34 3.1 1.62 0.57 1.15 100.0 1.566 1.570±.002 99.9±0.1 1,2 HD 147084 A4 II/III 0.72 3.9 4.34 0.67 1.15 32.0 3.823 3.803±.005 31.8±0.1 3,4 HD 154445 B1V 0.42 3.15 3.73 0.558 0.95 90.1 3.576 3.629±.010 89.4±0.1 1,2,3 HD 160529 A2 Ia 1.29 3.1 7.31 0.543 1.15 20.4 7.139 7.143±.022 20.8±0.1 1,2 HD 187929 F6-G0 I 0.18 3.1 1.76 0.56 1.15 93.8 1.696 1.684±.006 92.9±0.1 1

References: 1. Serkowski et al. (1975) 2. Hsu & Breger (1982), 3. Wilking et al. (1980), 4. Martin, Clayton & Wolff (1999)

strument. The stars observed and the parameters needed for our bandpass model are listed in table 3.

These stars are distant stars polarized by the interstellar medium. The wavelength dependence of interstellar polarization can be represented by the empirical model (Serkowski, Mathewson & Ford 1975; Wilking et al. 1980)

P (λ) = Pmax exp ( −K ln^2 λ λmax ) (3) where the values of Pmax, λmax and K are empirical parameters that are fitted to observed wavelength dependent polarization measurements and are given in table 3 for our standard stars. Since the wavelength dependence is well determined we can calculate the polarization we expect to observe with Mini-HIPPI by integrating over the passband as described by Bailey et al. (2015). The predicted polarizations are given in table 3.

We observed HD 84810 on 7 occasions, HD 147084 twice, and the other stars once. The average results are given in table 3. It can be seen that the agreement between predicted and observed polarization is excellent. The largest discrepancy between observed and predicted is 0.05% for HD 154445. Other objects agree to 0.01 – 0.03%. This is as good as can be expected considering the accuracy of the past results we are comparing with. The results demonstrate the accuracy of the instrument and of the bandpass model we are using to determine the efficiency corrections.

The position angles all agree with previous values to within one degree.

3.3 Comparison with other high precision polarimeters

In table 4 we compare Mini-HIPPI observations of a number of stars, most observed on multiple nights, with previous high-precision observations of these stars. Some of these earlier observations are with HIPPI on the Anglo-Australian Telescope, either from Cotton et al. (2016a), or from new observations first reported here in the case of ϵ Sgr and Acrux. Other observations are with PlanetPol on the William Herschel Telescope (Bailey, Lucas & Hough 2010). In making these comparisons it is important to note that the wavelengths are not the same. The HIPPI observations are in a narrower band, the SDSS g^′ band from about 400 – 550 nm, and the PlanetPol observations are in a much redder band from 590 – 1000 nm. The first three stars listed are stars that we use as low polarization calibrators, so it is not surprising that the measured polarizations are low. The next three stars (Fomalhaut, Arcturus and α Cen) are nearby bright stars that we are evaluating as potential bright low polarization calibrators. Canopus is a somewhat more distant object, that shows polarization, probably interstellar in origin. Acrux and ϵ Sgr (Cotton et al. 2016b) are B star systems that likely have both intrinsic and interstellar polarization contributions. It can be seen that the agreement between the Mini-HIPPI observations and earlier observations is typically at the ∼10 ppm level or better. Where there are somewhat larger differences these are entirely consistent with plausible effects due to wavelength dependence.

At low levels of polarization where the polarization is of a similar magnitude to the error, observed values of polarization are subject to a positive bias (Simmons & Stewart 1985). Polarization data are sometimes corrected by applying a debiasing correction of which one common form is to replace P by √ P^2 − σ^2_P . Following our procedure in earlier publications (Bailey et al. 2010; Cotton et al. 2016a) we have not applied any debiasing correction to the data tabulated here. We apply such corrections only when they are needed for specific statistical analyses where the bias becomes important. We also note that in most cases the problem caused by this bias can be avoided by working in normalized Stokes parameters (Q/I and U/I) rather than in P and θ.

It can be seen from the results in table 4 that Mini- HIPPI can easily detect polarization at levels ∼100 ppm as shown by the results on Canopus and ϵ Sgr, which agree very well in polarization and position angle with previous observations. In principle detection of polarization levels down to ∼ 20 − 30 ppm should be possible, but this is hard to test with the limited existing data on polarization at these levels.

It should be noted that Arcturus is reported by Kemp et al. (1986) to be a polarimetric variable with an amplitude of about 50 ppm in the B band and a possible period of around 45 days. Our current observations are insufficient to confirm such variability.

Our observations of α Cen apply to the combined light of both α Cen A and B. Although α Cen was previously observed by HIPPI (Cotton et al. 2016a) the observation is considered unreliable because α Cen B would have been close to the edge of the aperture.

3.4 Performance versus Magnitude

In figure 6 we have plotted the polarization errors derived from the data reduction system for a range of stars observed with Mini-HIPPI against the B magntitude of the stars. The measured errors have been scaled to a uniform integration

MNRAS 000, 1–7 (2015)

6 Bailey et al.

Table 4. Observations of low polarization stars compared with HIPPI and PlanetPol observations

Star HD Distance N obs Mini-HIPPI (350-700 nm) HIPPI (400-550nm) PlanetPol (590-1000nm) (pc) P (ppm) θ (deg) P (ppm) θ (deg) P (ppm) θ (deg)

Sirius 48915 2.6 11 3.1 ± 2.7 8 ± 25 5.5 ± 1.7 114 ± 18 β Hyi 2151 7.5 4 14.8 ± 11.5 127 ± 23 8.8 ± 2.5 95 ± 16 β Leo 102647 11.1 7 11.8 ± 7.8 21 ± 19 2.3 ± 1.1 35 ± 14 Fomalhaut 216956 7.7 6 17.8 ± 5.9 73 ± 10 24.3 ± 3.1 111 ± 7 Arcturus 124897 11.3 6 17.9 ± 4.4 7 ± 7 6.3 ± 1.6 31 ± 7 α Cen 128620/1 1.3 26 15.1 ± 1.9 34 ± 4 Canopus 45348 94.8 6 122.8 ± 4.0 115 ± 1 112.8 ± 1.7 116 ± 1 Acrux 108248 98.7 1 327.9 ± 14.1 114 ± 1 363.2 ± 1.7 113 ± 1 ϵ Sgr 169022 43.9 2 112.7 ± 11.4 34 ± 3 115.3 ± 1.9 37 ± 1

Figure 6. Polarization errors of Mini-HIPPI observations scaled to 1000 seconds integration and plotted against B-magnitude. The line has the slope expected for photon-noise limited performance and is fitted through the points for the fainter stars. The dashed line is the expected performance for Mini-HIPPI on a 1 metre telescope.

time (T ) of 1000 sec under the assumption that the error scales as T^{−0.5}. The actual integration times ranged from 400 to 1600 seconds. Over most of the range it can be seen that the error scales with magnitude in the way expected for photon shot noise limited performance as shown by the line on the diagram. However, the performance is relatively poorer (above the line) for the brightest stars. This could be an indication that the 500 Hz modulation we are using is not fast enough to completely remove scintillation noise for the brightest stars. It may also be in part due to the requirement to turn down the PMT gain by using a lower HT voltage to avoid saturation for these stars. The dashed line on the diagram shows the performance expected for Mini-HIPPI on a 1 metre telescope.

Figure 7. Mini-HIPPI observations (in black) of the polarization and position angle of Spica plotted against phase on the orbital period. The red points are observations of Spica in the g^′ band taken with HIPPI on the Anglo-Australian Telescope.

3.5 Observations of Spica

As an example of the type of science that can be done with Mini-HIPPI we present observations of Spica (α Vir). Spica is a binary system consisting of two B stars orbiting with a period of just over 4 days. Previous polarimetry of Spica was presented by Pfeiffer & Koch (1977) who described it as unpolarized and gave a value of P = 0.03± 0.01 %. Tinbergen (1982) reported observations showing a marginal detection of polarization in the Q Stokes parameter. Elias, Koch & Pfeiffer (2008) also report a small polarization in an average of 11 measurements but say that the standard deviation indicates that this star is not a polarization variable. Cotton et al. (2016a) presented three measurements of Spica with HIPPI on the AAT that showed clear polarization and significant variability.

In figure 7 we show observations of Spica obtained with Mini-HIPPI that show clear periodic variability in polarization and position angle. The observations are plotted against

MNRAS 000, 1–7 (2015)

High Precision Polarimeter 7

phase on the oprbital cycle according to the ephemeris of Tkachenko et al. (2016). Zero phase corresponds to the time of periastron passage.

Also plotted on figure 7 are eight observations of Spica obtained with HIPPI on the AAT in the g^′ band. Although this is a different band to the one used for Mini-HIPPI the effective wavelengths are similar. It can be seen that the HIPPI and Mini-HIPPI observations appear quite consistent.

These results show that Mini-HIPPI, even on a very small telescope, is capable of easily detecting polarization variation, where that was not detectable with a previous generation of polarimeters.

We are continuing to monitor the polarization of Spica with the aim of filling in remaining gaps in the phase curve, and we will present a full analysis at a later date. Spica is an important system, because it enables the polarimetric modelling of a binary system to be compared with the results of interferometric observations (Herbison-Evans et al., 1971).

4 CONCLUSIONS

We have described the Mini-HIPPI polarimeter, a compact and lightweight instrument based on the use of a ferroelectric liquid crystal modulator. Test observations with Mini-HIPPI on a 35 cm telescope show polarizations for standard stars in excellent agreement with predicted values. We measure the telescope polarization to be ∼60 – 90 ppm. A number of stars observed with Mini-HIPPI show polarization results typically within about 10 ppm or better of previous measurements with other high precision polarimeters. We have used Mini-HIPPI to study the polarization of the binary star system Spica, and it easily detects periodic polarization variations around the binary cycle which were not detectable with the precision available on earlier instruments.

ACKNOWLEDGEMENTS

This work was supported by the Australian Research Council through Discovery Projects grant DP 160103231.

REFERENCES

Bailey, J., Lucas, P.W., Hough, J.H., 2010, MNRAS, 405, 2570 Bailey, J., Kedziora-Chudczer, L., Cotton, D.V., Bott, K., Hough, J.H., Lucas, P.W., 2015, MNRAS, 449, 3064 Bott, K., Bailey, J., Kedziora-Chudczer, L., Cotton, D.V., Lucas, P.W., Marshall, J.P., Hough, J.H., 2016, MNRAS, 459, L109 Castelli, F., & Kurucz, R.L., 2004, arXiv:astro-ph/0405087 Cotton, D.V., Bailey, J., Kedziora-Chudczer, L., Bott, K., Lucas, P.W., Hough, J.H., Marshall, J.P., 2016a, MNRAS, 455, 1607 Cotton, D.V., Marshall, J.P., Bott, K., Kedziora-Chudczer, L., Bailey, J., 2016b, in Short, W., Caprarelli, G., eds, Proc. 15th Australian Space Science Conference, National Space Society of Australia, Sydney, p. 55 de Juan Ovelar, M., Diamantopoulou, S., Roelfsema, R., van Werkhoven, T., Snik, F., Keller, C., 2012, in Angeli, G.Z., Dierickx, P., eds, Proc SPIE Conf. Ser. Vol 8449, Modelling, Systems Engineering and Project Management for Astronomy V. SPIE, Bellingham, art. id. 844912 Elias, N.M., Koch, R.H., Pfeiffer, R.J., 2008, A&A, 489, 911

Gisler, D., Feller, A., Gandorfer, A., 2003, in Fineschi, S., ed., Proc SPIE Conf Ser. Vol. 4843, Polarimetry in Astronomy, SPIE, Bellingham, p 45 Herbison-Evans, D., Hanbury-Brown, R., Davis, J., Allen, L.R., 1971, MNRAS, 151, 161 Hough J.H., Lucas P.W., Bailey J.A., Tamura M., Hirst E., Harrison D., Bartholomew-Biggs M, 2006, PASP, 118, 1302 Hsu, J., & Breger, M., 1982, ApJ, 262, 732 Kemp, J.C., 1969, JOSA, 59, 950 Kemp, J.C. & Barbour, M.S., 1981, PASP, 93, 521 Kemp, J.C., Henson, G.D., Kraus, D.J., Beardsley, I.S., Carroll, L.C., Duncan, D.K., ApJ, 301, L35.

Kemp, J.C., Henson, G.D., Steiner, C.T., Powell, E.R., 1987, Nature, 326, 270 Lucas, P.W., Hough, J.H., Bailey, J.A., Tamura, M., Hirst, E., Harrison, D., 2009, MNRAS, 393, 229 Marshall, J.P. et al., 2016, ApJ, 825, 124 Martin, P.G., Clayton, G.C., & Wolff, M.J., 1999, ApJ, 510, 905 Pfeiffer, R.J., & Koch, R.H., 1977, PASP, 89, 147 Sabatke, D.S., Descour, M.R., Dereniak, E.L., Sweatt, W.C., Kemme, S.A., Phipps, G.S., 2000, Opt. Lett., 25, 802 Simmons, J.F.L., & Stewart, B.G., 1985, A&A, 142, 100 Serkowski, K., Mathewson, D.S., & Ford, V.L., 1975, ApJ, 196, 261 Tkachenko, A., et al., 2016, MNRAS, 458, 1964 Tinbergen, J., 1982, A&A, 105, 53.

Wiktorowicz, S.J. & Matthews, K., 2008, PASP, 120, 1282 Wiktorowicz, S., & Nofi, L.A., 2015, ApJ, 800, L1 Wiktorowicz, S., Nofi, L.A., Daniel, J.-T., Kopparla, P., Laughlin, G.P., Hermis, N., Yung, Y.L., Swain, M.R., 2015, ApJ, 813, 48 Wilking, B.A., Lebofsky, M.J., Martin, P.G., Rieke, G.H., Kemp, J.C., 1980, ApJ, 235, 905

This paper has been typeset from a TEX/L^ATEX file prepared by the author.

MNRAS 000, 1–7 (2015)